This device is a portable meter. It is capable of measuring coating thickness quickly, precisely and without injury, both for the laboratory and engineering environment. It is currently widely used in manufacturing, metal processing industry, chemical industry, commodity inspection and testing fields; and is essential for material protection major.
This device meets the following criteria:
GB/T 4956─1985 non-magnetic coating thickness measurement on magnetic metal substrate, Magnetic Method
GB/T 4957─1985 non-conductive coating thickness measurement on non-magnetic metal substrate, Eddy Current Method
JB/T 8393-1996 Magnetic&Eddy Current coating thickness measurement
The device uses magnetic thickness measurement method, which non-destructively measures thickness of non-magnetic coating (such as: aluminum, chromium, copper, enamel, rubber, paint, etc.) on magnetic metal substrates (such as steel, iron, alloy and hard magnetic steel, etc.).
Magnetic Method (F-type probe)
When the probe contacts the cover layer, the probe and magnetic substrate forms a closed magnetic circuit; Due to the presence of non-magnetic coating, magnetic resistance changes. The thickness of the cover layer can be derived by measuring the change.
Eddy Current Method (NF-type probe)
When the probe contacts the cover layer, the probe and non-magnetic substrate forms Eddy Current and gives feedback to the coil inside the probe. The thickness of the cover layer can be derived by measuring the feedback.